Payment & Shipping Terms:
|1.2M Cable Tie:||Used To Fix The Probe||Working Temperature:||-30℃~+75℃|
|Measurement Accuracy:||±0.5%;||Inbreaking Protection:||IP66|
|Peripheral Interface:||RS232 / RS485 / Analog Transmission||Length:||40cm-150cm|
battery gps tracker,
gps tracker with fuel sensor
0~5V output [CA-201A]:
Black line connect ground wire
Red line connect positive power
White line 0~5V voltage signal output,connect voltage signal wire
Yellow line empty fuel tank use only
Green line full fuel tank use only
This liquid level sensor is level measuring instrument,contain high resolution ratio and measurement accuracy.It is automatic calibration which does not need manual intervention,no temperature drift, and is not affected by medium change.In other words,regardless of the measured medium is water or gas or diesel, no matter how temperature changes, it can output signal accurate liquid level height at one sensor. Completely solve the problem of measure medium difficulty ,like ethanol and methanol,and the problem of measurement error caused by oil’s label and great temperature difference in different areas.
The equipment can be used cooperatively with GPS vehicle positioning terminal and other terminal to make vehicle positioning and fuel consumption management system.It can evaluate the usage of the vehicle and the driver's driving level,monitor gas up situation of vehicle,precise calculate hundred kilometers fuel consumption,completely eradicate the phenomenon of oil stolen,oil spilling,detouring,make extra money,private use of government vehicles,falsely making receive..
Examination area: 100~1000mm
resolution ratio: 0.1%
Maximal error of measure :1mm
Output signal :4~20mA,0~5V, RS485, RS232, CAN bus
Voltage supply :DC7~42V
Rated supply :0.1W
Fixed type :Screw installation M20X1.5 OR Flange mounting
Pressure range :-0.1MPa ~ 0.1 MPa
ambient temperature -40~85℃
2.1 definition of the output line
Interface output line use 6 core cable, connecting definition:
This sensor adopt high-speed MCU of CNC technology, adopting self-calibration
Calculation , and leaves out the complicated Manual calibration while using. in normal
circumstance, the sensor can directly be used in the measurement of the diesel, if need to calibration, you can use the three following ways to complete. (each can work)
1.self-calibration: put the sensor slowly into the The tested medium while power on, let the liquid get into the feed inlet slowly ,stoping when the liquid is overflew the vent hole of the sensor. and this operation accomplish the calibration of the sensor . In order to prevent the failure of the calibration process , please do this process for at least 2 times .( notice:This way is under the condition of powering on , outage restart will restore the factory Settings.In the condition of empty fuel tank,refueling to full, the sensor itself are also able to complete the self-calibration )
Manual calibration : at the time of empty fuel tank, put the yellow line short circuit with the ground wire( black line) , and then power on the sensor, operating for 60s or longer, after that , disconnect the back ground wire and the yellow line , and this operation could accomplish the calibration ( notice: Sensor already setted to 0 # diesel prior to shipment ,without Special Circumstances)
remote automatic calibration :if adopting RS232 or RS485 for data transmission, it can be accomplished the calibration by vehicle GPS location system sending orders. Sensor has two orders of calibration ,D1,D2. When the fuel tank empty , it will send D1, and when the fuel tank is full, it will send D2, and this operation could accomplish the remote calibration.
2.3 Installation Notes
Before installation, please check the attachment: Flange, rubber blanket,O-rings, and Whether the screw is consistent and complete
Trape the o-rings to the root of the sensor ( packed prior to shipment )
Daub the Oil-resistant sealant on the two side of the rubber pad, and aim flange plate to the tank flange hole,then fixed by screw,symmetrical tighten the strength in turn to make sure uniform stress and avoid oil spilling.
To finish the installation ,insert the sensor and use a wrench to tight.
connect the power cord, communication line, cable, ground wire according to the definition, and yellow line and green line need to trape one by one , preventing from the short circuit
Link the power of the sensor after checking the circuit
Installation position should be placed as near as possible to the center of the fuel tank, in order to reduce the influence of the oil level of tilt and the oil level of volatility while on the hill.
1.Only one signal can exist among RS232/ RS485/ 0-5V/ 4~20mA, and you can use either one of them. The trunk of RS485 can meet multiple sensors at the same time, but you should set the address of each sensor in advance. Address of each sensor must be unique, in case the trunk is locked. RS485 trunk is made of half-duplex chips, so when it reading data, you should set aside the corresponding time for the module. The reading speed cannot be too fast: when it reads continuously, the time interval between the two instructions cannot be less than five seconds.
2.For the signal transmission line, especially the one of RS232,communications circuits should not be too long, and should use the shielded wire.
3. For the sensors of the signal RS232/ RS485, when debugging the failure of communication, you can exchange the white line and the blue one and test again. If it works then, it is the connection error before. The operation is only for digital communications.
4. The power supply voltage of sensor cannot be more than the normal working voltage, and the power consumption of the sensor’s power supply cannot be less than the normal working power consumption.
5. The actual environment should conform to the requirements of the basic parameters of sensors, and shall not exceed the normal range, or it would damage the sensor, even cause other accidents.
6. On account of oil products, there are differences between the mediums of the calibration and use, users had better operate according to 2.2.2.
7. Do not remove sensors or damage it maliciously, or we do not guarantee it.
8.General factory accessories include Flange, rubber blanket, O-Rings, booklet of instruction, certificate of conformity. If you have special needs, please tell us in advance.
9.Guarantee period of the sensor is one year. All accessories including rubber blanket, O-Rings, are beyond the scope of “ threeguarantees ”.
Symptom of failure
1No-output /communication error
2The output value unchanged
3Measurement error of output value
4Output value floats up or down
1 Open circuit of the power or voltage too low
Wrong connection of the lines and wires
2 service voltage abnormal
3 service voltage abnormal
4 oil level flow
service voltage abnormal
Exclusion and solutions
1test connect→test if the voltage is greater than 10V when load→test the supply current [range 3~10mA] → connect method see 2.1,or change over blue and white line[digital communication only]→if all above is regular or current beyond normal range,the sensor occurred failure.
2Test the supply voltage when load[ex.voltage unstable] → recharge the sensor →test if there is a serious interference around→test whether sensors hole is blockage from top to bottom → execute 2.2.2→If not,the sensor occurred failure.
3 detect whether service voltage is stable while loading ( if the voltage is unstable) → →whether external have interference→check whether the sensor is jammed → execute the 2.2.2 process →then it should be the sensor fault
4 check the oil leve is stable → check the service is normal and functional →whether external have interference →execute the 2.2.2 process→then it should be the sensor fault
1 4~20mA output supply voltage DC20~28V,current is 4~20mA except communication output
2Check the tank oil level is float or not,if become regular after recharge,check the interference and the power
3 it usually can be functional while executing the 2.2.2 process
4 it usually occur on 0~5V signal
The ultrasonic fuel level sensor is used for digital recording vehicle refueling, to prevent driver stealing oil , avoid the waste of resources, improve operational efficiency, and improve the traffic safety, strengthen the operation and management level.
This sensor is applicable to various types of vehicles such as trailers, taxis, buses, concrete mixer truck, engineering machinery, sanitation trucks and other.
1. All of our product is free to use the Tracking service server in 24 hours each day
2. All of our product there is a 2 year quanlity assurance.
3. Sample fee will not be returned.
1. Standard configuration
1) main device 1 pcs
2) power line 1 pcs
5) Installation cables 1 set
6) Usage instructions
7) Warranty card
Contact Person: Jason