Payment & Shipping Terms:
|Product Name:||Fuel Sensor GPS Tracker||Type:||Capacitive Fuel Sensor|
|Examination Area:||100~1000mm||Resolution Ratio:||0.1%|
|Maximal Error Of Measure:||1mm||Heat-resistant:||-40~150|
|Voltage Supply:||DC7~42V||Rated Supply:||0.1W|
|Pressure Range:||-0.1 MPa To 0.1 MPa||Ambient Temperature:||-40~85℃|
|Output Signal:||4~20mA,0~5V, RS485, RS232, CAN Bus|
High Sensitivity Capacitive Fuel Sensor combined Capacitor Sensor And Liquid Level Sensor
CLS2-700 Capacitive Fuel Sensor combined capacitor sensor and liquid level sensor. This Product
is not only contain the features of high precision measurement and maintenance-free.
At the same time it does not influence by temperature and medium. It can be automatically measuring
error caused by different temperature or medium for precise compensation, achieving the purpose of
This liquid level sensor is level measuring instrument, contain high resolution ratio and measurement
accuracy. It is automatic calibration which does not need manual intervention, no medium is water or gas or
diesel, no matter how temperature changes, it can output signal accurate liquid level height at one sensor.
Completely solve the problem of measure medium difficulty, like ethanol and methanol, and the problem
of measurement error caused by oil's label and great temperature difference in different areas.
Before installation,please check the attachment: Flange,rubber blanket,O-rings, and whether the screw
is consistent and complete.
Trape the o-rings to the root of the sensor (packed prior to shipment)
Daub the Oil-resistant sealant on the two side of the rubber pad, and aim flange plate to the tank flange
hole, then fixed by screw,symmetrical tighten the strength in turn to make sure uniform stress and avoid
To finish the installation,insert the sensor and use a wrench to tight.
Connect the power cord, communication line, cable, ground wire according to the definition, and
yellow line and green line need to trape one by one, preventing from the short circuit.
Link the power of the sensor after checking the circuit.
Installation position should be placed as near as possible to the center of the fuel tank,in order to
reduce the influence of the oil level of tilt and the oil level of volatility while on the hill.
Tel: +86 18665990450
Contact Person: Arthur QI